VER3.20 - Feb 01, 2013
VER3.12 - Dec 10, 2012
VER3.10 - Nov 12, 2012
Changes from Ver. 3.12
The expression for calculating the measured power factor has been changed from a signed expression to an unsigned expression.
(λ=P/S → λ=|P|/S)
When the power is turned off, the panel’s OUTPUT OFF indication and the output status sometimes did not match. This has been corrected.
Changes from Ver. 3.10
The reliability in the internal communication of the parallel operation driver (PD05M-PCR-LE/PD05S-PCR-LE) has been improved.
The load meter full scale calculation when the current limit is set close to the minimum value has been corrected.
Changes from VER 2.00
Support for the parallel operation driver option
Support for the analog signal interface option
|New Feature||The 2P Mode On/Off function (for selecting the output mode) has been added to the AC mode that uses the single-phase, three-wire output driver option (2P05-PCR-LE).|
A communication interface command for querying the measured AC power (overall reactive power) has been added. (for the single-phase, three-wire output and three-phase output options)
|Improvement||When panel operation is not possible due to the current condition or conflicts with other settings, the reason is now indicated for better operability.|
|Bugfix||The reversed UVP/OVP judgment polarity on the V phase side when the single-phase, three-wire output driver option (2P05-PCR-LE) is in use has been fixed.|
|Bugfix||The overload judgment point (threshold) when the frequency is set less than 40 Hz in AC mode has been corrected.|
|Bugfix||The error in the calculation of the reactive power of the V and W phases when the single-phase, three-wire output or three-phase output option is in use has been fixed.|