Sets the trigger source for clearing sequence start trigger wait states.
You can select from the available choices below.
|Value||IMMediate||Executes the measurement values when the Initiate key on the PLZ-5W front panel is pressed or when INIT:ACQ is received from a PC or the like.|
|BUS||Records the measurement values at the timing of software trigger (*TRG) application. The trigger is applied to both trigger systems: (1) sequence and step (TRANsient) and (2) measure (ACQuire).|
|DIGITAL2||Executes the measurement values at the timing at which a signal is input to pin 13 of the EXT CONT connector of the PLZ-5W (when SYST:CONF:DIGITAL2:DIR is set to INPut).|
|TALink||If PROG:STEP<n>:TRIG:GEN is set to TALink, measured values are recorded during the step execution of the sequence.|
|MSYNc||Executes the step when TRIG:ACQ:MSYN is received from a PC or the like.|
Returns the source of the trigger of the ACQuire subsystem in character format in response to the TRIG:ACQ:SOUR? query.