Command (function search)
Synchronized operation is possible among multiple synchronized PLA-5Ws.
For details on how to establish synchronized connections, see the user’s manual contained in the accompanying CD-ROM.
Synchronizing load on/off
|INP:MSYN:ACC||Sets whether to accept load on/off sync signal (IMP:MSYN) input from other synchronized devices|
|INP:MSYN||Turns the load on or off simultaneously on synchronized PLZ-5Ws|
|TRIG:ACQ:SOUR||Measurement trigger source. To synchronize, specify MSYNc.|
|INIT:ACQ||Switches to measurement trigger wait state|
|TRIG:ACQ:MSYN||Starts measurement simultaneously on synchronized PLZ-5Ws|
Synchronizing the start of sequences
|TRIG:TRAN:SOUR||Sequence trigger source. To synchronize, specify MSYNc.|
|INIT:TRAN:PROG||Switches to the sequence trigger wait state in the program selected by PROG|
|TRIG:TRAN:MSYN||Starts a sequence simultaneously on synchronized PLZ-5Ws|
Synchronizing the resuming of sequences
|PROG:STEP<n>:TRIG:WAIT||Sets trigger wait at the specified step of the program selected by PROG and sets the trigger source To synchronize, set the trigger source to MSYNc.|
|TRIG:TRAN:MSYN||If synchronized PLZ-5Ws are in a MSYNc trigger wait state in a program step, sending this command to any of the PLZ-5Ws starts the step simultaneously on the synchronized PLZ-5Ws.|