This document is prepared for the Kikusui Easy Controller for PCR-M application, VER 1.4.0 (JUNE 2012)
To run the application, the following operational environment is required:
To install this app software, run ezCtrlPcrmSetup_1_x_x.exe (x may differ depending on version) under Administrator priviledge. By installing the app, the IVI-COM instrument driver for PCR-M series will also be installed.
This application communicates with your PCR-M AC Power Supply through the GPIB, USB, or RS232 interface, therefore you must connect the instrument to your PC with an appropriate cable. Plus you must set up the instrument CONFIG settings, by pressing the CONFIG button on the instrument front panel.
Mind that the instrument must be restarted (turn OFF then ON) to take effect the configuration changes, if you have changed any.
This application is an interactive virtual front panel program, which remote-controls your PCR-M AC Power Supply. The application operates not only for most of front panel controls, it also provides file I/O capabilities to SAVE/OPEN panel settings, and provides measurement data logging storing in a text file.
Immediately after the application is launched, the I/O Config dialog appears to confirm the I/O connection with your instrument. Then, click the "Try *IDN" button to check the communication. If the instrument identity string could be returned, it will be shown. If it is failed, you need check the I/O cable, VISA and IVI driver setup, and instrument configurations once again. By checking the Enable Simulation checkbox, you can operate as a simulation without connecting actual instrument. (The simulated operation assumes that a PCR500M is virtually connected.) By clicking the OK button, the application's main screen will appear.
You can remote-control your instrument regarding "Coupling Mode", "Voltage Range", "Output Settings", "Limit Settings", etc....
To turn ON or OFF the output, click the "Output ON" or "Output OFF" button. Mind that some settings cannot be changed while the output is ON. A large red display on the scrren "Output ON" or "Output OFF" is automatically refreshed so that it can indicate the correct instrument status, however it works only if the "Status Monitor" is enabled (default at application startup is "enabled"). If the "Status Monitor" is disabled, that large indicator is not automatically refreshed because the application does not monitor the actual instrument status in this case.
Clicking the "Status" button enables/disables the auto-refresh feature of the "Status Monitor" indicators. The default state at the application startup is "enabled".
Clicking the "Measure" button enables/disables the auto-logging feature of the "Measurement Monitor" list. If the "Data Generation" option (by operating Tools | Data Options menu) is configured for Generate Measurement Files enabled, the auto-logging for measurement data will be stored in a text file placed in the specified directory. The file name is automatically assigned according to the present date/time. Each text file contains a tab-separated list so that you can open it with Microsoft Excel or with other spread-sheet applications
Clicking the "Pk Clear" button clears the peak-hold data for the current measurement.
Clicking the "Alm Clear" button clears the instrument Alarm currently occurring. The "Alarm" includes any items shown in the "Status Monitor" area in red colour. Mind that DSP alarm, if generated, requires the instrument to be restarted.
Clicking the "Recall" or "Store" button recalls/stores the panel setting memory.
Clicking the "Open" or "Save" button opens/saves the panel setting from/to file system. The default file name extension is .PNL and the format is the Windows .INI file style text.
By selecting View | Refresh (F5) menu, you can refresh the application's GUI so that the screen contents synchronise with the actual instrument settings. This is useful for GUI recovery when you have operated the instrument front panel at LOCAL state thereby the consistency has got corrupted.
By selecting Instrument | Enable Continuous Measure menu, you can enforce the instrument to automatically refresh the voltmeter/ammeter displays. As the normal instrument behaviour in REMOTE state is that the voltmeter/ammeter are refreshed *only when* the explicit measurement action is initiated. Therefore, the instrument's static operation in REMOTE without any measurement command to be sent, keeps the voltmeter/ammeter unchanged. To avoid this, you can enable the Continuous Measure feature, so that the instrument automatically refreshes the voltmeter/ammeter displays without having to receive a measurement initiation command, as if it were operating in LOCAL state.
This application handles instrument errors by employing the "Error Query" feature, so that the application can be simpler design. Any instrument errors, such as setting "Ext AC" Coupling Mode without installing an "External Board" option, will be hooked up whenever you do erroneous operations.
When the "Status Monitor" indicates the DSP event highlighted, this means one or more internal errors are generated. In this case, you can identify error details by the Self Test and Error Query features. By selecting Tools | Self Test menu, you can perform the self test. If the self test result is non-zero, then operate Error Query by selecting Tools | Error Query menu.
This application controls the instrument through the IVI instrument driver. The driver equips the "State Cache" feature that increases the I/O performance by avoiding redundant settings to be sent multiple times. For example, once you have set the AC Voltage 110V, subsequent operations for the same settings do not generate I/O traffic. However, if you have operated the instrument front front panel by switching it to the LOCAL state, the instrument driver and the application do not know that the instrument settings were altered, therefore the driver's state cache assumption is now corrupted. To avoid this problem, you can temporarily "invalidate" the state cache giving the application/driver to have an opportunity to synchronise with the instrument settings once again.
VER 1.4.0 - 19/JUNE/2012, Bundled IVI driver changed to 184.108.40.206 (x86/x64).
VER 1.3.0 - 10/MAY/2010, Architecture changed due to bandling new IVI-COM driver. Windows Vista and 7 support added, Windows 9x/Me support removed.
VER 1.2.0 - 15/MAY/2006, Frequency item is added on Monitor and Log
VER 1.1.0 - 06/DEC/2005, Memory Recall Preview function added
VER 1.0.1 - 13/JULY/2005, Simulation function added
VER 1.0.0 - 01/JUNE/2005, the first release
(c)2005-2012 Kikusui Electronics Corp.