Aborts operations such as setting, change, and measurement in all sequence groups.
The trigger status immediately after the power is turned on is the same as the trigger status when the ABOR command is received. If the ABOR command is sent when the measurement is already started, the measured data of the sequence 3 group remains invalid.
If the ABOR command is sent when the sequence 3 group is not initiated and the measured data that is held is valid, the measured data is not discarded.
A specific sequence group cannot be specified with the ABOR command. It is always interpreted as ALL.