A device driver supporting USB T&M Class (USBTMC) is required to control the PWX through the USB interface. The USBTMC driver is automatically installed by the VISA library.
By changing the legacy language and emulation settings, you can execute existing programs that were written for other instruments. Normally, select the SCPI language and turn emulation off (these are the factory default settings).
Use a standard USB cable to connect the PWX to the computer.
1. In the CONFIG settings, set the emulation (CF22).
Normally, use the default value (CF22: nonE).
2. Restart the PWX.
The PWX is equipped with service request and serial polling functions.
Complies with USB Specification 2.0
Complies with USBTMC Specification 1.0 and USBTMC-USB488 Specification 1.0
Data rate: 480 Mbps maximum (High speed)
VID （Vender ID）
PID （Product ID）
750W model: 0x1025
1500W model: 0x1024